Socket Meter Test Bench ANSI Standard
|FOB Price:||US $15,000-45,000 / Piece|
|Min. Order:||1 Piece|
|Min. Order||FOB Price|
|1 Piece||US $15,000-45,000/ Piece|
|Payment Terms:||L/C, T/T, D/P, Western Union, Paypal, Money Gram|
- Energy Meter Test Equipments
- 27 Products
- Product Image
- Product Name
- FOB Price
- Product Package
- Product Attribute
- Product Table
- Model NO.: SP-6153
- Maxcapacity: 0-200A
- Load Way: Electronic Load
- Display: Digital
- Weight: 100-500Kg
- Reference Standard: Sp-5300
- Transport Package: Woodencase
- Origin: China
- Type: ANSI Three Phase
- Accuracy Grade: 0.05%
- Loading Method: Dynamic Load
- Control: Computer Control
- Power Source: AC110V
- Trademark: POWOO
- Specification: IEC
- HS Code: 9028301900
SP-6153 three-phase ANSI Energy meter stationary test bench is designed for testing three-phase energy meters accuracy below 0.2, especially for testing Closed I-P Links three-phase energy meters. Test bench can generate 3-phase voltage, 3-phase current and any power factor signal for tested meter, the reference meter's accuracy is class 0.05(0.1).
a. DDS technology signal generator:
The signal generator adopts digital sine wave composing technology. The signal generator can trace the voltage phase and supply a dummy load to the tested meter by outputting adjustable current with different phase;
b. High efficiency D-Class current amplifier and voltage amplifier:
The equipment use PWM power amplifier(or we called D-Class amplifier), which is high efficiency (>85%) and very low heat consumption. The amplifier has protection circuit itself, it can avoid damage due to wrong connection or other manual operations. Current amplifier and shifter can output any current from 0.001A to 100A; Voltage amplifier can output any voltage from 50V to 450V;
c. DSP+MCU dual processor designed multifunction reference meter.
The reference meter use one DSP to sample the signal of three phase voltage and three phase current, it sample the voltage waveform and current waveform continuously, even there are harmonic exist in the waveform, the reference meter can also measure the voltage/current/power in dependable status; And other MCU is designed to display the tested result and receive input information.
d. DCS framework designed error calculator system.
The error calculator system is designed as DCS framework, every meter position has one isolated error calculator, error data is transferred from RS485 Bus, PC can receive every meter's error data and then displayed it on the screen, the error calculator can receive signal which due to TTL / OC or Optical head output.
e. Can be controlled from keyboard or software.
The equipment has operated keyboard itself, the equipment can be operated form software and also it can be operated from keyboard. Without PC you can also test meter's error.