Volumetric Water Meter Test Bench
|FOB Price:||Get Latest Price|
|Min. Order:||1 Piece|
|Payment Terms:||L/C, T/T, D/P, Western Union, Paypal, Money Gram|
- Water Meter Test Bench
- 12 Products
- Product Image
- Product Name
- FOB Price
- Product Package
- Product Attribute
- Product Table
- Model NO.: WV20-DN15-25mm
- Maxcapacity: Dn25mm
- Load Way: Mechanical Load
- Display: Digital
- Weight: 500-1000Kg
- Reference Standard: Volumetric Standard Vessel
- Transport Package: Woodencase
- Origin: Shanghai China
- Type: Test and Measurement Instruments
- Accuracy Grade: 0.2
- Loading Method: Dynamic Load
- Control: Manual Control
- Power Source: AC380V
- Trademark: POWOO
- Specification: ISO4064
- HS Code: 9028201000
The water meter testing bench is the necessary measure instrument for the test and repair of water meter in metrological service, Water meter manufacture, Water supply Company, industrial and mining establishments.
The tester bench is made of stainless steel materials; The surface is treated with polishing.
Clamp equipment and drain machine is controlled by four-way reversing valve; The water flow rate is adjusted by copper case stainless steel core ball valve, it has high accuracy, the structure is compaction, long life, easy operate and maintain.
2. Main technical parameter
2.1 The tester bench accuracy: 0.2
2.2 Tested water meter size: DIN15, DIN20, DIN25.
2.3 Technical criteria for water meter testing: Used for full capability test or produce and maintenance
2.4 Instantaneous flow index gauge is adopted glass rotor flowmeter group, the actual
Measurement accuracy not less than 2.5%
The detail type and specification as following:
2.5 Working volume container cubage:
15-25mm, (10L+10L)20L, (50L+50L)100L,
2.6 Using water source working pressure: 0.25-0.4MPa
2.7 Clamp equipment flex distance is:
2.8 Appearance size
3600× 1650× 2100 (in series meter tester)
3 Structure and working principle
The water meter tester bench is composed by work desk, clamp equipment, front and back straight pipe, flowmeter, standard volume container, drain water valve.
After the water from the steady sources come into the enter place of tester bench, the water will come into the tested meter in straight pipe, and then come into the choosed standard volume container, compare the reading number between tested meter and standard volume container, and get the error of testing water meter.